I wanted to use this driver , win 7 but said there was no driver in this directory. In windows 7 you can simply reboot and then press F8 as Windows begins to reboot to get the windows boot menu. If there are any unnecessary devices plugged into other ports then remove them and try again. Use your cursor keys to highlight the option ‘Disable Driver Signature Enforecement’ then then hit the enter key to continue booting. Next click the ‘Details’ tab and in the drop-down box select ‘Hardware Ids’. For windows 7 you will find in the C:
|Date Added:||26 November 2004|
|File Size:||41.38 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Right click on it and select ‘Properties’.
cy7c68103 It will normally be listed as ‘Unknown Device 1’. In the window that opens up under the ‘General’ tab and check to see if there are any errors reported. Second wire of switch may to connect to R87, that near 24C This cannot be undone.
If you installed the driver more than once you may need to repeat this step to remove every version.
HC CY7CA 16CH Logic Analyser (HCTEST) –
But I’ve not managed to get any of them to help: Thank you for all your work to support this Radio. If you see any errors try reinstalling the driver but first remove the currently installed driver by: Probably R in zero position now, that SWR 1,0: If it fails qin7 appear check the following: Next hit the restart button.
If possible try a different USB cable. If you downloaded the Sigrok files in the above step to your home folder then the required file will be in a location similar to this: You can not post a blank message. This will automatically install a driver for the analyser.
After installing SuiteUSB 3. To do this you can use the Zadig driver install win which is downloaded with the Sigrok Pulseview software: I am attaching a inf file for 64bit win 7if you want for 32bit then modify the vid pid under ;for x86 platforms.
New program for the CY7C
When I disconnected wim7 it still reads 1. A latest windows build of the PulseView software can be downloaded here: The new topic will begin with this message. Unable to claim USB interface.
I am resistant to upgrade but many have with varying degrees of success. Subject of the new topic: Even when connected to an antenna that is not tuned it will show 1. Do you know if anyone has tried this with Windows 10?
Welcome to the Cypress Developer Community 3. Author Post time Subject Direction: Versions of Ubuntu prior to To use your analyser with the Sigrok software you will first need to install a driver.
If you purchased your analyser after this date then please try the following driver first: The latest version of the Sigrok software can be downloaded from the cy7cc68013 Sigrok website sigrok.
When used with the sigrok PulseView software you can capture and decode up to 16 digital channels at a maximum sample rate of 12MHz. I will try this new firmware shortly.
Windows has stopped this device because it has reported problems.